MPI530 meter is a multifunctional instrument for electric shock protection measurements including the option of various methods for measuring earthing and earthing resistance. Additionally the meter enables its user to analyse power supply parameters, records and displays them on its screen. The instrument automatically measures the insulation resistance in a socket and in 3 , 4  and 5wire cables when used with AutoISO1000C adapter. Measurement results may be continuously verified, depending on user’s demand or according to binding regulations. Memory of the instrument may store tens of thousands of measurements allowing users to store them in a structure of a protocol and describe individual measurement points.
INTERACTIVE PRESENTATION OF SONEL MPI530
Measurement of short circuit loop impedance:
 impedance measurement with resolution 0,001 Ω and 23 A current (44 A phasetophase)  shortcircuit resistor R _{ ZW } =10 Ω,
 measurement range: 95...440 V, frequency 45...65 Hz,
 measurement of short circuit loop impedance with resolution 0,01 Ω, in protected systems without tripping the RCD’s with I _{ Δn } ≥ 30 mA,
 automatic calculation of shortcircuit current, differentiating between phasetophase and phasetoneutral voltage,
 measurements using UNISchuko plug with measurement triggering button (also when L and N leads are exchangeable) or 1,2 m, 5 m, 10 m, 20 m test leads, with optional use of 3phase socket adapters (AGT),
 selection of installation protections and automatic evaluation of measurements results.
Tests of residual current devices (RCD), types AC, A, F, B and B+:
 measurement of general, shorttime delay and selective RCD’s with rated residual current of 10, 30, 100, 300, 500 and 1000 mA,
 automatic measurement of all RCD parameters (when the START button is pressed, the meter performs the whole measurement cycle, including the.
LPE short circuit loop impedance measurement with 15 mA current),
 user selected waveform of forced leakage current: sinusoidal (start from increasing or decreasing edge), unidirectional pulsating (positive or negative), unidirectional pulsating with DC bias (positive and negative), direct (positive and negative),
 measurement of tripping current I _{ A } using the ramp current,
 measurement of tripping time t _{ A } at ½I _{ Δn } , 1I _{ Δn } , 2I _{ Δn } and 5I _{ Δn } ,
 measurement of touch voltage UB and protective conductor resistance RE without the RCD tripping,
 detection that L and N conductors are switched in the socket; will not affect the measurement,
 measurement of tripping current I _{ A } and actual tripping time t _{ AI } at single activation of RCD,
 measurement for 95...270 V voltage.
Insulation resistance measurements:
 test voltage values: 50 V, 100 V, 250 V, 500 V and 1000 V,
 insulation resistance measurements up to 10 GΩ,
 insocket measurements using the UNISchuko adapter,
 meter is protected against voltage on the tested object and voltage appearing during the measurement,
 autodischarge of the object after the measurement,
 automatic measurement of all resistance combinations in three, four and five core leads using the additional AutoISO1000C adapter,
 acoustic signalling of fivesecond periods to enable determining the time profile during the insulation resistance measurements.
Lowvoltage resistance measurement of equipotential and protective bonding:
 continuity measurement of protective conductor with ≥200 mA current in both directions,
 low current measurement with sound signal,
 autocalibration of test leads – leads of any length can be used.
Earth resistance measurements:
 measurement with 3 and 4pole methods with 2 additional electrodes,
 measurement with the 3pole method with and additional clamp,
 measurement with 2clamp method.
Display range  Resolution  Accuracy 
0,000...19,999 Ω  0,001 Ω  ±(5% m.v. + 0,03 Ω) 
20,00...199,99 Ω  0,01 Ω  ±(5% m.v. +0,3 Ω) 
200,0...1999,9 Ω  0,1 Ω  ±(5% m.v. + 3 Ω) 
Display range  Resolution  Accuracy 
0,00...19,99 Ω  0,01 Ω  ±(6% w.m. + 10 digits) 
20,0...199,9 Ω  0,1 Ω  ±(6% w.m. + 5 digits) 
200...1999 Ω  1 Ω 
Measurement of earth resistance RE with the 3p and 4p method
 0,50 Ω…1,99 kΩ for test voltage 50 V
 0,56 Ω…1,99 kΩ for test voltage 25 V
Display range  Resolution  Accuracy 
0,00...9,99 Ω  0,01 Ω  ±(2% m.v. + 4 digits) 
10,0...99,9 Ω  0,1 Ω  ±(2% m.v.. + 3 digits) 
100...999 Ω  1 Ω  
1,00...1,99 kΩ  0,01 kΩ 
 test voltage: 25 V or 50 V rms
 test current: 20 mA, sinusoidal rms 125 Hz (for f _{ n } =50 Hz) and 150 Hz (for f _{ n } =60 Hz)
 measurement blocked at interference voltage U _{ N } > 24 V
 maximum measured interference voltage U _{ nmax } = 100 V
 maximum resistance of auxiliary earth electrodes 50 kΩ
Selective earth resistance measurement with clamp (3p + clamp)
Display range  Resolution  Accuracy 
0,00...9,99 Ω  0,01 Ω  ±(8% m.v. + 4 digits) 
10,0...99,9 Ω  0,1 Ω  
100...999 Ω  1 Ω  
1,00...1,99 kΩ  0,01 kΩ 
 measurement with additional current clamp
 interference current measuring range: up to 9,99 A
Selective earth measurement with two clamps
Display range  Resolution  Accuracy 
0,00...9,99 Ω  0,01 Ω  ±(10% m.v. + 4 digits) 
10,0...99,9 Ω  0,1 Ω  
100...999 Ω  ±(20% m.v. + 4 digits) 
 measurement with transmitting and receiving clamps
 interference current measuring range: up to 9,99 A
Soil resistivity mesurement (ρ)
Display range  Resolution  Accuracy 
0,0...99,9 Ωm  0,1 Ωm 
Depending on accuracy
of R _{ E } measurement 
100...999 Ωm  1 Ωm  
1,00...9,99 Ωm  0,01 kΩm  
10,0...99,9 kΩm  0,1 kΩm 
 measurement with Wenner’s method
 distance settable in metres or feet
 distance range: 1...30 m (1...90 feet)
Phase sequence indication
 Phase sequence indication: conforming, nonconforming
 Mains voltage range ULL: 100...440 V (45...65 Hz)
 Display of phasetophase voltage values
RCD type  Current  Range  Resolution  Accuracy 
General and
shorttime delay 
0,5*I _{ Δn }  0...300 ms 
1 ms 
± (2% m.v. + 2 digits)
(for RCD with I
_{
Δn
}
=10 mA

1*I _{ Δn }  
2*I _{ Δn }  0...150 ms  
5*I _{ Δn }  0...40 ms  
Selective  0,5*I _{ Δn }  0...500 ms  
1*I _{ Δn }  
2*I _{ Δn }  0...200 ms  
5*I _{ Δn }  0...150 ms 
accuracy of residual current application: for 0,5*I _{ Δn } :8...0% for 1*I _{ Δn } , 2*I _{ Δn } , 5*I _{ Δn } : 0...8%
Measurement of RCD tripping current IA for sinusoidal residual current (AC type)
Rated
current 
Meas. range  Resolution  Test current  Accuracy 
10 mA  3,3...10,0 mA  0,1 mA  0,3 x I _{ Δ } ..1,0 x I _{ Δ }  ± 5%I _{ Δn } 
30 mA  9,0...30,0 mA  
100 mA  33...100 mA  1 mA  
300 mA  90...300 mA  
500 mA  150...500 mA  
1000 mA  330...1000 mA 
The measurement can be started from positive or negative halfperiod of forced leakage current
Rated
current 
Meas. range  Resolution  Test current  Accuracy 
10 mA  3,5...20,0 mA  0,1 mA  0,35 x I _{ Δn } ..2,0 x I _{ Δn }  ±10%I _{ Δn } 
30 mA  10,5...42,0 mA  0,35 x I _{ Δn } ..1,4 x I _{ Δn }  
100 mA  35,0...140 mA  1 mA  
300 mA  105...420 mA  
500 mA  175...700 mA 
Measurement for positive or negative halfperiods of forced leakage current
Measurement of RCD tripping current IA for direct residual current (type B)
Rated
current 
Meas. range  Resolution  Test current  Accuracy 
10 mA  2,0...20,0 mA  0,1 mA  0,4 x I _{ Δn } ..2,0 x I _{ Δn }  ±10%I _{ Δn } 
30 mA  6...60 mA  1 mA  
100 mA  20...200 mA  
300 mA  60...600 mA  
500 mA  100...1000 mA 
 for U _{ N } = 50 V: 50 kΩ...250 MΩ
 for U _{ N } = 100 V: 100 kΩ...500 MΩ
 for U _{ N } = 250 V: 250 kΩ...999 MΩ
 for U _{ N } = 500 V: 500 kΩ...2 GΩ
 for U _{ N } = 1000 V: 1,000 MΩ...9,99 GΩ
Display range*)  Resolution  Accuracy 
0...1999 kΩ  1 kΩ  ±(3% m.v. + 8 digits) 
2,00...19,99 MΩ  0,01 MΩ  
20,0...199,9 MΩ  0,1 MΩ  
200...999 MΩ  1 MΩ  
1,00...9,99 MΩ  0,01 GΩ  ±(4% m.v. + 6 digits) 
*) not greater than measuring range for given voltage
Range  Resolution  Accuracy 
0,00...19,99 Ω  0,01 Ω  ±(2% m.v. + 3 digits) 
20,0...199,9 Ω  0,1 Ω  
200...400 Ω  1 Ω 
 voltage on open terminals: 4…9 V
 output current at R < 2 Ω: min. 200 mA
 autocalibration of test leads
 measurements for both current polarities
Illuminance measurement
Range  Resolution  Accuracy 
0,1...99,9 lx  0,1 lx  ±(5% m.v. + 2 digits) 
100...999 lx  1 lx  
1,00...9,99 klx  0,01 lx  
10,0...19,9 klx  0,1 lx 
  measurement in luxes (lx) or feetcandles (fc)
ANALYSIS AND RECORDING OF SINGLEPHASE SYSTEM
 Voltage measurement ULN: 0...500 V
 Frequency range of measured voltages: 45,0...65,0 Hz
 Frequency measurement range for 50...500 V voltages: 45,0...65,0 Hz (basic uncertainty max. ± 0,1% m.v. + 1 digit)
 cosφ measurement: 0,00...1,00 (resolution 0,01)
 Measurement and recording in singlephase system
Current measurement (True RMS)
Clamps  Range  Resolution  Accuracy * 
C3, C6  0,0...99,9 mA  0,1 mA  ±(5% m.v. + 3 digits) 
100...999 mA  1 mA  
C3,C6
F1, F2, F3 
1,00...9,99 A  0,01 A 
±(5% m.v. + 5 digits)
(C3, C6) ±(0,1% In + 2 digits) (F1, F2, F3) 
10,0...99,9 A  0,1 A  
100...999 A  1 A  
F1, F2, F3  1,00...3,00 kA  0,01 kA 
* the accuracy of current clamps must also be taken into account
Measurement of active power P, reactive power Q, apparent power S and cosφ
Range
[W], [VA], [var] 
Resolution
[W], [VA], [var] 
Accuracy * 
0...999  1  _{ ±(7% m.v. + 3 digits) } 
1,00...9,99 k  0,01 k  
10,0...99,9 k  0,1 k  ±(7% m.v. + 5 digits) 
100...999 k  1 k  
1,00...1,50 M  0,01 M 
 voltage range: 0…500 V
 current range: 0...1000 A (3000 A)
 mains rated frequency fn: 50 Hz, 60 Hz
Voltage harmonics measurement
Range  Resolution  Nr harm.  Accuracy 
0,0...500 V  0,1 (1*) V  1,2,...15  ±(5% m.v. + 3 digits) 
16,...40  ±(5% m.v. + 10 digits) 
 in addition display of h02...h40 values as percent of h01 (up to 999%)
*) from 300 V to 500 V
THD (in relation to the 1 st harmonics)
Resolution  Accuracy  
THDF voltage
(h = 2…40) 
0,0…999,9%
for U _{ RMS } ≥ 1% U _{ nom } 
0,1%  ±5% 
THDF current
(h = 2...40) 
0,0…999,9%
for I _{ RMS } ≥ 1% I _{ nom } 
0,1%  ±5% 
NOTE: “m.v.” means “ measured value”
The instruments conforms to:
 EN 61010  1 (general safety requirements)
 EN 61010  031 (detailed safety requirements)
 EN 61326 (electromagnetic compatibility)
 EN 61557  10 (requirements for combined instruments)
 IEC 60364  661 / HD 60364  6 (measurements  verification)
 IEC 60364  441 / HD 60364  441 (measurements – protection against electric shock)
 EN 04700 (measurements – acceptance tests)
 EN 12464 (lighting of work places)
 DIN VDE 0100
 DIN VDE 0413
 BS 7671
Complete device information  
Manufacturer site: 
Sonel MPI530 Multifunction Electrical Installations Meter
 Brand: Sonel S.A.
 Model: MPI530
 Product Number: WMGBMPI530
 Availability: 1 to 7 days order processing

0.00€
Tags: Sonel, MPI530, Multifunction, Electrical, Installations, Meter